Cable Fault Finder and Length Measurement TDR
Accurately finds cable length, impediments in the cable and conditions at the end of every wire in your data, power, or communications/video system up to 3,000 ft.
With the complexity of today’s electrical, data, voice and video signal carrying systems, you need a cable fault locator time domain reflectometer tool that is fast, accurate and easy to use to see down the wire and show the exact conditions that exist. Snap Shot™ is that tool. Made in America with American Spread Spectrum Time Domain Reflectometry technology, Snap Shot™ is the new 21st century tool for cable fault location for every type of cable in any industry. Advanced Spread Spectrum TDR technology works in all wire environments and is immune to voltage, current, or data stream disruptions. See the true picture of your wiring and cable systems with the Snap Shot™ cable fault detector and length measurement tool today.
Spec Sheet: Snap Shot™ Specification Sheet
User Manual: Snap Shot™ User Manual
|Snap Shot Warranty Registration Giveaway: FREE T3 Hanging Pouch|
|Whitepaper: Spread Spectrum Time Domain Reflectometry (SSTDR)|
Snap Shot™ Demo Video
Q. What is SSTDR?
A. SSTDR stands for Spread Spectrum Time Domain Reflectometry. It differs from traditional TDR measurement devices by using multiple spread spectrum signals.
Q. Can The Snap Shot measure a single conductor wire?
A. No, TDR and SSTDR require two conductors to measure a wire or cable
Q. How do I turn off the unit?
A. Hold down the Snap Shot power button for 3 seconds to power the unit off
Q. How do I calibrate the NVP?
A. Power on the Snap Shot. Get a known length of cable that is 25ft or longer and connect it to the Snap Shot. Press the CAL button and use the up/down buttons until you see the length of the cable. Press the Test button.
Q. How do I save the NVP?
A. The Snap Shot has the ability to save two NVP values. Press the MEM button to toggle between M1 and M2. To save the current NVP value, select the memory slot you wish to use and press and hold the MEM button. The arrows to the left of the NVP value will disappear when the value has been saved.
Q. Can the Snap Shot save test results?
A. No, the Snap Shot does not save test results. If you would like this feature see T3 Innovation's Arc Chaser (link to AC page)
Q. Will the Snap Shot display the voltage on the line?
A. No, the Snap Shot does not have functions of a Digital Multi Meter. If you would like this feature see T3 Innovation's Power Prowler (link to AC page)
Q. How do I use the tone generator?
A. The tone generator places an AM frequency on the cable that the Snap Shot is connected to. A tone probe (sold separately) picks up the frequency and produces an audible sound. You can trace the tone to follow the cable to an open or a break.
Q. Why do cable measurements like Romex differ in length using the Snap Shot when coiled vs. uncoiled?
A. The cable impedance can be affected by coiling because it creates inconsistency between the spacing of the conductors which can affect the NVP value.
Q. Do I have to press the test button every time I want to run a test?
A. No, you can activate the loop test mode by holding the “Test” button down for three seconds.
Q. Does the Snap Shot have a backlight?
A. Yes, tap the power button to turn the backlight on and off.
Q. Can I measure a cable that is connected to a device like an IP camera or a breaker?
A. In most cases, yes. If the device has open/closed circuitry, the Snap Shot will be able to measure to that circuit. If a resistor, transistor, capacitor etc. is in the path of the cable and the device circuit, the Snap Shot may not be able to obtain an accurate measurement.
Q. Does the Snap Shot test underground cable?
A. The Snap Shot can measure underground cable as long as it’s insulated and has two conductors.
Q. Will the Snap Shot be damaged if I connect to a cable with voltage?
A. The Snap Shot does have voltage protection but should not be connected to voltage over 60 volts due to the risk of electrical shock.
Q. Why does the Snap Shot display Err when I am trying to measure a cable?
A-1 The cable is too long.
A-2 The cable has excessive attenuation.
A-3 The NVP is not correct.
A-4 The calibration length is not close to the length of the cable.
Q. What is the warranty on the Snap Shot?
A. One year from date of purchase.
|CA001||Cable Assembly, Insulated Push on F to RJ45 Plug|
Cable Assembly, Insulated Push-On F-Conn to Alligator Clips
|AD001||F-Connector Coupler, High Performance F81|
|AD002||Adapter, F-Jack to BNC Jack|
|CP100||Hanging T3 Pouch|